For the best experience on the Abcam website please upgrade to a modern browser such as Google Chrome
Take a look at our BETA site and see what we’ve done so far.
Search and browse selected products
Purchase these through your usual distributor
The quickest way to estimate the amount of protein in solution is to use UV-vis to measure absorbance directly, but this is generally not very accurate or sensitive. Highly accurate quantitation of most proteins can be achieved using either a Bradford or bicinchoninic acid (BCA) assay.
The table below summarizes the main differences between these different methods.
|UV-vis / A280||Bradford assay||BCA assay|
|Accuracy||Low; usually used as an approximation||High; gives accurate quantitation||High; gives accurate quantitation|
|Speed||Very fast; < 1 min||Fast; 20 mins||1 hr, 30 mins|
Tryptophan and Tyrosine residues absorb at 280 nm
Binding of Bradford reagent to protein; complex absorbs at 595 nm
Proteins reduce Cu2+ to Cu+; BCA chelates Cu+; complex absorbs at 562 nm
|Considerations||DNA/RNA can interfere with results if present in high quantities; nucleic acids absorb at 260 nm|
Will not detect peptides without Tryptophan and Tyrosine amino acids
SDS (>0.1 %) and other detergents should be avoided
EDTA (>10 mM) and other strong chelating agents should be avoided
Reducing agents should be avoided
Protein concentration can be estimated by measuring the UV absorbance at 280 nm; proteins show a strong peak here due to absorbance from Tryptophan and Tyrosine residues (commonly referred to as A280). This can readily be converted into the protein concentration using the Beer-Lambert law (see equation below). This method is used mostly for a very rough estimation of concentration, e.g. to check the success of purification.